I am not sure maybe something missing during installing. 0 Kudos Message 7 of 13 (5,742 Views) Reply 0 Kudos Re: Error -88302 occurred at DAQmx Create Channel (AI-Voltage-Basic).vi:1 Error -88302 ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://jcl.error.963.pcfixspeed1.com/ Unable to determine IP address from host name jcl.error.963.pcfixspeed1.com The DNS Depending on your version of LabVIEW, one easy first step would be to reinstall your DAQmx driver or upgrade to the latest version supported by your version of LabVIEW. Test Panels works well as before(which works well before i reinstalled the DAQmx driver), but still have problem "Error -88302 occurred at DAQmx Control Task.vi:3". http://zeronews.net/ni-error/ni-error-91.html
This is common error code format used by windows and other windows compatible software and driver vendors. Primary Software: Primary Software Version: 4.6.1 Primary Software Fixed Version: N/A Secondary Software: LabVIEW SignalExpress>>Development System Problem: When I try to run a DAQmx Acquire step in SignalExpress or the self Thanks for your help! I've attached the screen shots as well as the vi. https://forums.ni.com/t5/LabVIEW/Error-88302-occurred-at-DAQmx-Create-Channel-AI-Voltage-Basic-vi/td-p/1585462
All the above actives may result in the deletion or corruption of the entries in the windows system files. Thnaks 0 Kudos Message 4 of 13 (5,780 Views) Reply 0 Kudos Re: Error -88302 occurred at DAQmx Create Channel (AI-Voltage-Basic).vi:1 Error -88302 o… Member 06-07-2011 03:33 AM Options Mark as DAQmx Create Channel polymorphic VI - what to select when connecting ICP pressure sensor Error -200220 occurred at DAQmx Create Channel Error -200220 occurred at DAQmx Create Channel Possible Bug: DAQmx
If I open the test panel for the last 3 indented devices, I can read voltage signals from each one. Regards,Austin S.National InstrumentsAcademic Field Engineer 0 Kudos Message 6 of 13 (5,755 Views) Reply 0 Kudos Re: Error -88302 occurred at DAQmx Create Channel (AI-Voltage-Basic).vi:1 Error -88302 o… Member 06-08-2011 01:48 You mentioned that you did the self-test and it passed, but did you try using Test Panels and were you able to see your signal as expected? Ni Error 88302 Error Codes are caused in one way or another by misconfigured system files in your windows operating system.
This article contains information that shows you how to fix Ni Error 88302 both (manually) and (automatically) , In addition, this article will help you troubleshoot some common error messages related My system was working without issue, my computer was unplugged unintentionally and now it will not work anymore. Showing results for Search instead for Did you mean: Reply Topic Options Start Document Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the http://forums.ni.com/t5/LabVIEW/Error-88302-occurred-at-DAQmx-Create-Channel-AI-Voltage-Basic-vi/td-p/3200140 I have a cdaq 9139 with 2 NI 9234 AI modules and 2 NI 9264 AO modules and also a TCP 2212 touch panel.
Please tell us why. this contact form Please Contact NI for all product and support inquiries. Let me know if anyone has run into this issue before and has fixed it, or if there is a workaround. Computer: HP Elitiebook 6390p Modules: 9472, 9221 Steps anyone who can help me?
I have looked for the solutions in the forum but I can't get to the configuration settings as mentioned in this solution: http://digital.ni.com/public.nsf/websearch/19E7993CF8366EED8625764200771C68?opendocument&Submitted&&node=133020_USIs there any other ways to solve the error?DAQ Assistant.gif 84 KB voltage.png 41 KB 0 Kudos Message 9 of 13 (5,731 Views) Reply 0 Kudos Re: Error -88302 occurred at DAQmx Create Channel (AI-Voltage-Basic).vi:1 Austin_S Member 06-08-2011 06:09 Maybe something wrong with LV? have a peek here Regards, Lauren Applications Engineer National Instruments 0 Kudos Message 2 of 4 (630 Views) Reply 0 Kudos Re: Error -88302 occurred at DAQmx Create Channel (AI-Voltage-Basic).vi:1780001 cjb6970 Member 10-07-2015 10:39
The Ni Error 88302 error is the Hexadecimal format of the error caused. Follow the steps below to cure this problem. Runs first time but on next iteration it returns error when trying to Register for Event. Error details: Error -200949 occurred at Register For Events Possible reason(s): You
I have used the channels in RSE mode and LabVIEW version is 2012. Thanks0 0 11/26/14--14:56: Error -200077 occurred at DAQ Assistant Contact us about this article Published (2016-06-14 06:15:00) How do you create a channel property in a TDMS file. I've done a "Self-Test" to each one and everything passes. United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Discussion Forums : Most Active Software Boards : LabVIEW : Error -88302 occurred at DAQmx Create Channel (AI-Voltage-Basic).vi:1
There is no activity on the Access Led of the scope card, but Windows detects thescope card. Multifunction DAQ Register for the community · Log in · Help CommunityCategoryBoardDocumentsUsers turn on suggestions Auto-suggest helps you quickly narrow down your Yes, the 9.2.3 is compatible with your version of LabVIEW, but make sure that you download the DAQmx 64 bit version driver (and it wouldn't hurt to go ahead and install Check This Out This thread profile page shows the thread statistics for: Total Authors, Total Thread Posts, and Thread Activity Home| About Us| Submit Your Site| Update Your Site| Get Search For Your Site|
I use DAQmx task and register 'Signal Event", when I am done I call Unregister for Events funciton and Stop DAQmx. Click here follow the steps to fix Ni Error 88302 and related errors. Worked a treat!” Colin- 1 Month Ago “Thanks for sharing, I no longer have to put up with the dreaded Ni Error 88302” Irvin- 2 Months Ago “Will this work with I've tried using the TDMS_SetChannelProperty as follows for (i = 0; i < numChannels; i++) error = TDMS_SetChannelProperty (channels[i], :Color, TDMS_UInt32, colors[i]); No error is generated but the
DC Voltage. You can specify a built-in property or a property created by TDMS_SetChannelProperty. There are two (2) ways to fix Ni Error 88302 Error: Advanced Computer User Solution (manual update): 1) Start your computer and log on as an administrator. 2) Click the I dont know why the second channel data appears in first channel, even though the voltage value connected is less on first channel (AI0) If I remove the second channel
That's wonderful news since it ensures that ni error 88302 chances are good your problem continues to be effectively documented and will more likely be sorted out by YOU! I can see it as a USB device under "Devices", but the Amber Light just stays on "ready" light. Generated Thu, 01 Dec 2016 16:12:55 GMT by s_wx1193 (squid/3.5.20) Regards,Austin S.National InstrumentsAcademic Field Engineer 0 Kudos Message 10 of 13 (5,713 Views) Reply 0 Kudos « Previous 1 2 Next » All Forum Topics Previous Topic Next Topic Privacy |
I've tried restarting my system, but to no avail. It's important to scan your PC every now and again to ensure that these files are in place and everything is as it should be. As a first step, I would strongly suggest repairing or uninstalling your DAQmx driver and reinstalling it. All rights reserved. | United States MY ACCOUNT   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Discussion Forums : Most Active Software Boards : LabVIEW : Error -88302 occurred at