You can no longer read any data. The device might need to be recalibrated or repaired by NI. You should update your device with the latest software. Error -200087Channel is not in the task, and the channel is not a valid global channel. http://zeronews.net/ni-error/ni-error-91.html
Make sure the spelling of the topology is correct and that the switch supports that topology. Replace the invalid character with a valid symbol or alphanumeric character. Error -200032Subset length specified is not valid. Error -200181Channel name specified is not valid for the switch device.
Find My Dealer © 2016 Vernier Software & Technology, LLC. Error -200056Channel name not specified in the list entry. Error -200044Route cannot be made between the source and destination terminals.
Right-click the item under Physical Channeland choose Change Physical Channel. Because of system and/or bus-bandwidth limitations, the driver could not write data to the device fast enough to keep up with the device output rate. To avoid this error, you can do any the following: 1. Error -200009Timestamps have been overwritten.
If you have a PXI chassis, identify the chassis correctly in MAX, and make sure it has been configured properly. To do so go to Control Panel»Device Manager and expand Data Acquisition Devices. Error -200223Specified threshold and hysteresis values for this channel create a triggering range that is not supported by your device. https://www.vernier.com/til/2148/ The settings requested by the most recent DAQmx Write will be applied on the next sample clock.
Error -200208Range statement in the list entry contains an invalid character sequence. Could you give me some suggestions to solve the problem? Showing results for Search instead for Did you mean: Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic to the Top Bookmark If the module is not visible in Device Manager or never appears in MAX, it could indicate that the device is damaged.
Error -200072The NI PXIe-5606 is not associated with a digitizer or an LO. http://188.8.131.52/t5/LabVIEW/error-201003-occurred-at-DAQmx-Create-Channel-AI-Temperature/td-p/1716500/jump-to/first-unread-message Error -200148FREQOUT counter cannot generate the desired frequency. Error -200168Number of channels to acquire exceeds the device maximum. Also, verify that the rate of the external clock matches the specified clock rate.
Error -200033DAQmx Write was invoked more than once between two consecutive sample clocks. this contact form Error -200222Acquisition has been stopped to prevent an input buffer overwrite. The identifier should specify a valid waveform or script name. Refer to documentation for information about the maximum length allowed for user-defined information.
Then I open Measurement & Automation explorer in order to locate the problem, but I find almost all the control menus of PCI-6220 are grey, including self-test , test panels,etc. You can also associate the NI PXIe-5603 with additional IF conditioning and RF conditioning modules. Use the high resolution clock, or increase your sample rate. have a peek here Error -200106Property must have the same value for all channels on this device.
Error -200076Current data supplied is outside of the specified range. You can also associate the NI PXIe-5605 with additional IF conditioning and RF conditioning modules. Device is powered, but was temporarily without power.
Error -200193Switch device does not support this operation. Refer to Measurement & Automation Explorer or your hardware documentation for valid terminal names. An attempt was made to create a handshaking task when the lines were previously reserved for static digital operations. Error -200226Trigger bus to which the device is connected does not have any free trigger lines for the driver to choose.
You must open the Configuration panel and associate the NI PXIe-5603 with a supported or external digitizer and a supported or external LO. Reset the device by doing one of the following: 1. Ensure that the reference signal applied is the same as the values that were input. 2. Check This Out Error -200047Invalid identifier after the device identifier in the list entry.
For generations, you can disable the regeneration of old samples.